BW GasAlertMicroClip XL - How to Manually Bump Test
In this guide, we’ll talk you through how to manually bump test your GasAlertMicroClip XL from BW Technologies.
Before bump testing, ensure you are using an approved gas and that the cylinder is not past its expiration date. Most manufacturers recommend that your gas monitor should be bumped daily but just make yourself aware of any site-specific or company-specific procedures around bump testing. Read our guide to bump testing gas detectors here.
Once you’ve switched it on the MicroClip XL will complete a self-check. After this has been completed it will then sit at a standard measurement mode.
You’ll need the calibration adapter which comes with the unit, calibration tubing, a known concentration of calibration gas, a suitable regulator and of course the monitor itself. In this case we’re using 25 ppm h2s, 100 ppm co, 2.2% ch4 and 18% o2 in a balance of Nitrogen.
You’re now ready to bump test your GasAlertMicroClip XL.
Attach the tubing to the adapter, which in turn is attached to the regulator and calibration gas bottle, attach the calibration clip to the monitor (the sensors of the GasAlertMicroclip XL are situated on the lower front part of the monitor), and then turn on the gas.
The sensors should start to respond and you should see values on the screen similar to those on your gas cylinder.